Santa Barbara Infrared’s MB Series blackbodies are absolute only temperature “set and forget” infrared calibration systems. They are designed to support single or multi non-uniformity correction test requirements. A user can quickly establish the blackbody temperature setpoint between 0°C and 90°C by using the included IRWindowsTM Lite software. This temperature setpoint value is stored into the system’s non-volatile memory, so no further PC connection is required after the initial temperature configuration. The operator can verify the blackbody is at the temperature setpoint by viewing the green “ready” indicator light on the back of the blackbody head. SBIR’s MB Series blackbody systems are designed for use as flood sources for non-uniformity correction (NUC) and for calibration of IR sensors.
NIST Traceable Calibration
Accuracy of +/-1.0ºC
Stability of +/-0.10ºC
A surface treatment is used to create a spectrally flat emissivity of > 0.970 (3µm to 5.5µm), > 0.950 (8µm to 14µm)
“Set & Forget” Programming
No Separate Controller Required
Below is a mechanical envelope for the MB Series blackbody system.
|Available Sizes||4” square aperture|
|Temperature Range||0ºC to 90ºC Absolute (at 20ºC ambient temperature)|
|Emissivity||> 0.970 (3µm to 5.5µm), > 0.950 (8µm to 14µm)|
|Uniformity||>96% of Tset – Tambient or 0.050ºC whichever is greater, over 90% of the emitting surface area|
|Approximate Heating Rate||+25ºC/min (at 25ºC)|
|Approximate Cooling Rate||-11ºC/min (at 25ºC)|
|Operating Temperature||10ºC to 40ºC|
|Storage Temperature||-20ºC to 70ºC|
|Relative Humidity||5% to 95%, non-condensing|
|Maximum Power Consumption||180W|
|Approx. Blackbody Weight||MB-04 – 5.3 lbs|
Notes: 1. Slew rates can vary depending on environmental conditions as well as initial blackbody temperature in relation to commanded temperature setpoint.
* Specifications subject to change without prior notice
Print PDF Datasheet
Please contact the SBIR sales team at (805) 965-3669 or via firstname.lastname@example.org to ensure the proper part number and to receive a quotation.
Information subject to change without notice.